1. Choose an acrylic glovebox for a laboratory
2. Internal equipment
If a balance needs to be placed inside, Model MA/MB/MC011 are MA/MB/MC001 are suggested. The dimension of the left port is 350mm in diameter.
3. Best value for price
MA/MB/MC012, MA/MB/MC011 and MA/MB/MC010.
4. Moisture and oxygen content
For MB with an antechamber to replace the gas, the humidity can be controlled at around 10%. As for MC models, the humidity can get to 1%-5% (about 100-300ppm). If lower moisture is needed, placing adsorbents such as molecular sieve or phosphorus pentoxide can be helpful.
5. Vacuum pump
Vacuum pumps with pumping speed between 4L-6L are recommended.
6. Furnace inside
The surface temperature of the furnace should not exceed 50 degree C.
7. Use of organic solvents inside?
Acrylic materials can be dissolved in most organic solvents. Use of organic solvent especially those volatile solvents may cause damage to the glovebox.
8. Use of Lithium Battery Electrolyte
A: The LIB electrolyte is corrosive, so it is not suggested to use it inside the acrylic glovebox, however, occasional use is all right, as long as there is no spill.
9. What is the difference between models MA, MB and MC?
A: Type MA is a simple isolation box, with the acrylic thickness of 10mm; Type MB adds an antechamber which can be vacuumed, so gas inside the antechamber can be replaced to keep low moisture and oxygen. Both the main chamber and antechamber in Type MC can be vacuumed. After vacuuming, high purity gas is injected to maintain very low moisture and oxygen inside.
10. Spin coating machine inside
If a spin coater is needed inside the glovebox, there must be a separate vacuum system for it. Please consult before placing an order.
11. Left port on the glovebox
The loft port can be custom made in square shape. But it is not recommended as the sealing is not as good as a round port.
In our products formaldehyde content is pretty low and meets health standard.
13. What should I pay attention while using the glovebox?
Type MA and MB glove boxes are simple. The gas inside can be directly replaced with inert gas to achieve the desired environment. As for Type MC glovebox, the glove ports must be sealed and vacuumed before vacuuming the main chamber, otherwise, the gas sealed in the gloves will inflate and explode the gloves. At the same time, the antechamber needs to be vacuumed before vacuuming the main chamber.
14. Use hygrometer inside the glovebox.
A hygrometer works when the moisture is above 20%. It doesn’t work for moisture below 20%. A dew-point meter is needed for Type MC glovebox.